Dräger Short Term Detection Tubes - Phosphoric Acid Esters 0.05/a (Pack of 10)6728461
Dräger Tubes For Short Term Measurements: Phosphoric Acid Esters 0.05/a
Dräger Safety offer a massive range of short-term gas detection tubes for the measurement of various gases. This particular set of tubes is for (Phosphoric Acid Esters 0.05/a).
The tubes come in a pack of 10 tubes and have a shelf life of 2 years.
|Standard Measuring Range:||0.05 ppm Dichlorvos|
|Number of Strokes n:||10|
|Time for Measurement:||approx. 5 min|
|Standard Deviation:||± 30 %|
|Colour Change:||yellow to red|
Ambient Operating Conditions:
|Temperature:||10 to 40 °C|
|Absolute Humidity||3 to 18 mg H2O / L|
a) (CH3O)2PO2-CH=CCI2 + Cholinesterase ~> inactive enzyme b) Butyrylcholine iodide + H2O ~> Butyric acid c) Butyric acid + Phenol red ~> yellow reaction product If phosphoric acid esters are present the enzyme is inactivated and butyric acid will not form, thus the weak alkali solution colors the indicating layer red and must be stable for 1min. If the enzyme remains active, phosphoric acid esters are not present, and the indicating layer remains yellow because of butyric acid formation.
Other phosphoric acid esters than dichlorvos are also indicated, but with different sensitivities.
After performing the required 10 pump strokes the reagent ampoule must be broken and the liquid transferred to the enzyme layer by gently tapping the side of the tube. The substrate layer must not become wet After waiting one (1) minute the liquid must be carefully drawn up to the marking line using the pump. Wait another minute before drawing the liquid onto the indication layer using the pump..
For more information on Dräger-Tubes click the image below.
To view our range of Dräger pumps please click below.
We Also Recommend